Despite increasing prevalence of T1D in ethnic minorities, they remain underrepresented in research studies evaluating diabetes technologies. Experiences of diverse cohorts are essential to assessing acceptability of diabetes devices. The Control-IQ Observational (CLIO) Study is an ongoing prospective, real-world evaluation of the Tandem Diabetes Care t:slim X2 insulin pump with Control-IQ technology in diverse people with T1D.
We evaluated glycemic metrics from adults completing the 12-month CLIO study. Participants with ≥75% overall CGM use during this time were included in the analysis. Outcomes were analyzed using a Wilcoxon test. Sample included 1,045 adults (921 Whites, 74 Hispanics, 34 Blacks, 16 Asians) with Black participants reporting the highest baseline HbA1c (median=8% [IQR=7-8.6)). Glucose Management Indicator (GMI) at study end reflected glycemic improvements for participants with Blacks showing a clinically relevant 0.7% overall reduction (7.3% [6.9-7.6]) (p<0.001) (Figure 1). Overall, sensor time in range (TIR) at 12 months ranged from 64.9% [56.8-70.5] for Black participants to 72.6% [62.8-80.8] for the White cohort. These findings confirm long-term successful use of Control-IQ technology across diverse ethnicities. Further evaluation of glycemic disparities is recommended to help inform patient support systems.
Watch the video from Steph Habif, VP Behavioral Sciences for a review of the presentation delivered at the conference.
Consult user guides for our pumps, infusion sets, predictive technology, and reporting app.View Documentation
Use our education tutorials to train your patients or CDEs on our easy-to-use products.View Resources
1. Breton MD, Kovatchev BP. One Year Real-World Use of the Control-IQ Advanced Hybrid Closed-Loop Technology. Diabetes Technol Ther. 2021;23(9):601-608.
2. Barnard K, et al. Patient-Reported Outcomes and Continuous Glucose Monitoring: Can we do better with artificial pancreas devices? Diab Care. 2015;38(5):e70.
3. Brown SA, Kovatchev BP, Raghinaru D, et al. Six-month randomized, multicenter trial of closed-loop control in type 1 diabetes. N Eng J Med. 2019;381(18):1701-1717.
4. Forlenza GP, Breton MD, Kovatchev BP. Candidate selection for Hybrid Closed Loop Systems. Diabetes Technol Ther. 2021;23(11):760-762.
5. Habif S, Singh H, et al. 217-OR: Glycemic Outcomes by Ethnicity in Adults with Type 1 Diabetes Using Control-IQ Technology: Early Results from the CLIO Study. Diabetes. 2021;70(Supplement_1):217-OR.
Important Safety Information
RX ONLY. The t:slim X2 pump and Control-IQ technology are intended for single patient use. The t:slim X2 pump and Control-IQ technology are indicated for use with U-100 insulin only.
t:slim X2 insulin pump : The t:slim X2 insulin pump with interoperable technology is an alternate controller enabled (ACE) pump that is intended for the subcutaneous delivery of insulin, at set and variable rates, for the management of diabetes mellitus in people requiring insulin. The pump is able to reliably and securely communicate with compatible, digitally connected devices, including automated insulin dosing software, to receive, execute, and confirm commands from these devices. The t:slim X2 pump is indicated for use in individuals six years of age and greater. Control-IQ technology: Control-IQ technology is intended for use with a compatible integrated continuous glucose monitor (iCGM, sold separately) and ACE pump to automatically increase, decrease, and suspend delivery of basal insulin based on iCGM readings and predicted glucose values. It can also deliver correction boluses when the glucose value is predicted to exceed a predefined threshold. Control-IQ technology is intended for the management of Type 1 diabetes mellitus in persons six years of age and greater.
|WARNING: Control-IQ technology should not be used by anyone under the age of six years old. It should also not be used in patients who require less than 10 units of insulin per day or who weigh less than 55 pounds.|
Control-IQ technology is not indicated for use in pregnant women, people on dialysis, or critically ill patients. Do not use Control-IQ technology if using hydroxyurea. Users of the t:slim X2 pump and Control-IQ technology must: use the insulin pump, CGM, and all other system components in accordance with their respective instructions for use; test blood glucose levels as recommended by their healthcare provider; demonstrate adequate carb-counting skills; maintain sufficient diabetes self-care skills; see healthcare provider(s) regularly; and have adequate vision and/or hearing to recognize all functions of the pump, including alerts, alarms, and reminders. The t:slim X2 pump, transmitter, and sensor must be removed before MRI, CT, or diathermy treatment. Visit www.tandemdiabetes.com/safetyinfo for additional important safety information.